The hostname associates a name with a numerical IP address. The unit is now ready to operate. This menu displays the list of supported hardware for the installed version of NI ID querying is discussed in the next section. The utility displays four options for resolving the verification error. One of the following should occur:.
|Date Added:||24 September 2004|
|File Size:||43.93 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Once your computer restarts to complete the installation of the NI Instruments must be powered on and connected to the GPIB controller to be recognized. The utility displays four options for resolving the verification error. Once you have determined the GPIB address of your instruments, it is easy to establish communications to verify that you can send and receive data to and from the instrument.
The Ethernet address is not the IP address.
GPIB-ENET/ – National Instruments
Exit when you are finished using the Ethernet Device Configuration utility. To communicate with your instrument at Address 4, follow these instructions:. The current hostname is displayed. ID querying is discussed in the next section.
Connect the other end of the Ethernet cable to your Ethernet network. This is indicated by an exclamation point! Connect one end of the power cord to the power supply.
National Instruments GPIB-ENET/ – In Stock, We Buy Sell Repair, Price Quote
Many times, systems are powered down and cables are disconnected from instruments for maintenance or system configuration changes.
It makes GPIB instrument detection and control easy by providing tools that help you search for connecting instruments, and send snet receive communication with your device. Alternatively, system power may be disrupted for a variety of reasons or the actual instrument may begin to malfunction.
If you are setting up your own network, you probably do not have a gateway or DNS server, so ebet should set these values to 0. Numbers between 0 and are valid for the third number of the IP address, but this number must be the same as other devices on your network if your subnet mask is The basic functions and concepts described above may seem simple; however, they can be snet in troubleshooting and getting your GPIB system up and running.
If you have already seen and closed the initial NI Configuration in Windows for USB. The utility alerts you if it detects a problem with the network settings.
With this test panel, you can set properties for your instrument communication as well as read and write to the instrument. View the glib for the device. Install Instrument Drivers Next. In addition to manually configuring the network parameters, you can use the Ethernet Device Configuration utility for any of the following purposes:.
With the VISAIC utility, you can speed 11000 application development by learning how to automate measurements with your instruments, uncover GPIB problems, and avoid headaches by identifying malfunctioning instruments.
Installing GPIB-ENET/100 Controllers
Run the Ethernet Device Configuration Utility. Using your computer, you can take advantage of this powerful development and debugging tool to interactively communicate read, write, serial poll, and so on with your GPIB instruments. Also have the network administrator assign the proper subnet mask, gateway, and DNS server addresses.